Diposable Medical Protective Clothing
Scope of Application
For clinical medical personnel in the work of contact with the potentially infectious patient's blood, body fluids, secretions etc. to provide barrier, protection.
1. Water permeability resistance
The hydrostatic pressure of the key parts of the protective clothing should not be lower than 1.67 Kpa (17cm H20)
2. Moisture permeability
The moisture permeability of protective clothing materials shall be no less than 2500g/(m2.d)
3. Resistance to synthetic blood penetration
The Resistance of protective clothing to synthetic blood penetration should be no less than 1.75 kPa.
4. Surface moisture resistance
The water level on the outer side of the protective clothing shall not be lower than grade 3.
5. Breaking strength
The breaking strength of materials in key parts of protective clothing shall be no less than 45N.
6. Elongation at break
The elongation at break of key parts of protective clothing should not be less than 15%.
7. Filtration efficiency
The filtration rate of non-oil particles in key parts and joints of protective clothing should not be less than 70%.
8. Antistatic property
The electric quantity of protective clothing should be no more than 0.6m/piece.
Model and Technological Parameters